INSTRUMETN CSSTSLS
GRATING GI
WAVELENGTH 6200 10000

# 1 order (BEAM A) *******************
BEAMA 496 1205
MMAG_EXTRACT_A 30
MMAG_MARK_A 30
# 
# Trace description 
#
DYDX_ORDER_A 1
DYDX_A_0 1.8079411230667526 0.000294687296425808 0.00013075268730441774 -5.8695979322444216e-08 -1.8530865927043794e-08 -3.4435334927576097e-10
DYDX_A_1 -0.007559325619461293 -2.235588199890071e-07 -4.861200082190469e-07 5.67597561284664e-11 9.665085002819538e-12 -5.654003972981209e-12
# 
# X and Y Offsets 
#
XOFF_A 0.0
YOFF_A 0.0
# 
# Dispersion solution 
#
DISP_ORDER_A 1
DLDP_A_0 357.25672649585346 0.0026817631051531373 -0.0008088235647495063 1.8090568199353523e-08 -1.8174081782777532e-07 4.848228169943458e-08
DLDP_A_1 8.722697564037938 0.000205426058276804 6.458898780582076e-07 5.754551977966249e-09 3.665174051543842e-10 -6.499751993083657e-11
#
SENSITIVITY_A GI.Throughput.1st.fits
#


# 0 order (BEAM B) *******************
BEAMB -121 80
MMAG_EXTRACT_B 30
MMAG_MARK_B 30
# 
# Trace description 
#
DYDX_ORDER_B 0
DYDX_B_0 -0.346400522896565 0.00018739376670127208 5.56857577173339e-05 -1.70474412603054e-08 -2.2972785159783426e-08 -3.6984395352510956e-09
# 
# X and Y Offsets 
#
XOFF_B 0.0
YOFF_B 0.0
# 
# Dispersion solution 
#
DISP_ORDER_B 1
DLDP_B_0 -69799.91747887168 -2.9724644701190073e-07 1.5072336267962783e-07 2.015287719689396e-11 1.0102759666168193e-11 -2.921355374946831e-11
DLDP_B_1 -3799.9960108170285 1.230476741723959e-08 5.120008047919456e-09 -1.7260127319496886e-12 -2.894859545656582e-13 -2.518720263725144e-13
#
SENSITIVITY_B GI.Throughput.0st.fits
#


# -1 order (BEAM C) *******************
BEAMC -1226 -531
MMAG_EXTRACT_C 30
MMAG_MARK_C 30
# 
# Trace description 
#
DYDX_ORDER_C 1
DYDX_C_0 2.077313161445304 0.000980949024127796 0.00021283436080014677 -2.6874752294266443e-07 1.2116296272703047e-08 -2.34247574816342e-08
DYDX_C_1 0.0006570356116067526 1.2290395976811375e-06 1.6637183470133416e-07 -3.0304924532261765e-10 2.4734040679617618e-11 -2.3590221649145147e-11
# 
# X and Y Offsets 
#
XOFF_C 0.0
YOFF_C 0.0
# 
# Dispersion solution 
#
DISP_ORDER_C 1
DLDP_C_0 -121.78907997722058 0.0032696940319881414 0.0006464378401977006 -7.640813428906632e-07 -1.8609591417862846e-07 8.290436333072529e-09
DLDP_C_1 -8.97979084681475 -0.00019598212441758928 3.783145685550695e-07 -6.5034895421998424e-09 -1.2413982090479435e-10 2.8899833209195562e-11
#
SENSITIVITY_C GI.Throughput.-1st.fits
#


# 2 order (BEAM D) *******************
BEAMD 1133 2430
MMAG_EXTRACT_D 30
MMAG_MARK_D 30
# 
# Trace description 
#
DYDX_ORDER_D 1
DYDX_D_0 6.774750256151258 4.707107703698734e-05 0.000503719215056276 -9.167623136460398e-09 -4.152231401897418e-08 -9.419809130154712e-09
DYDX_D_1 -0.011168343549873953 2.6666977679877337e-08 -7.470831119863737e-07 -6.286511915787003e-12 1.6439841503101933e-11 2.596573361763681e-12
# 
# X and Y Offsets 
#
XOFF_D 0.0
YOFF_D 0.0
# 
# Dispersion solution 
#
DISP_ORDER_D 1
DLDP_D_0 644.6040954256978 -0.006725435944267471 -0.00046759138288674147 6.074871380608643e-07 5.934218163422943e-08 4.8842737161317195e-08
DLDP_D_1 4.024665076158718 0.00010081087363835187 7.778605740527281e-08 2.3982243719693978e-09 -9.612044507219901e-12 -2.68791877244903e-11
#
SENSITIVITY_D GI.Throughput.2st.fits
#


# -2 order (BEAM E) *******************
BEAME -2388 -1152
MMAG_EXTRACT_E 30
MMAG_MARK_E 30
# 
# Trace description 
#
DYDX_ORDER_E 1
DYDX_E_0 7.575427899521311 -0.00021618386157363904 0.0005591063261388894 7.342842947533113e-09 -1.7485023350712322e-08 -1.0377783468068108e-08
DYDX_E_1 0.004683960947904331 -1.888624610879838e-08 3.2884821904527606e-07 1.018984346983241e-11 5.168946855257885e-12 -2.043053548121042e-12
# 
# X and Y Offsets 
#
XOFF_E 0.0
YOFF_E 0.0
# 
# Dispersion solution 
#
DISP_ORDER_E 1
DLDP_E_0 267.93286227255373 0.00201938462017863 0.0006580841742383252 1.342728670389483e-07 8.208055062958599e-08 -5.8696202469254026e-08
DLDP_E_1 -4.25924990707677 -9.333093346801208e-05 3.0998041792804064e-07 -2.5252108562599416e-09 6.108018499993683e-11 -2.776870248444526e-11
#
SENSITIVITY_E GI.Throughput.-2st.fits
#

