INSTRUMETN CSSTSLS
GRATING GU
WAVELENGTH 2550 4000

# 1 order (BEAM A) *******************
BEAMA -1079 -453
MMAG_EXTRACT_A 30
MMAG_MARK_A 30
# 
# Trace description 
#
DYDX_ORDER_A 1
DYDX_A_0 -1.92965371692521 -0.00014210305610185418 0.0002793831881261664 7.297873089403136e-09 1.5594886191413381e-09 -1.0034451568844385e-08
DYDX_A_1 -0.004128564707327051 1.3112855729371645e-09 2.9857740533775283e-07 -1.0633208432943827e-11 1.2571557443148134e-11 -6.047189646816668e-12
# 
# X and Y Offsets 
#
XOFF_A 0.0
YOFF_A 0.0
# 
# Dispersion solution 
#
DISP_ORDER_A 1
DLDP_A_0 108.37296206647136 -0.003058754198396685 -0.000903055915044194 2.9810529336266413e-07 -8.932479411361967e-09 9.889950193647952e-08
DLDP_A_1 -4.470739824905015 3.021295891394025e-05 -1.5666894728459118e-06 1.234812844316597e-10 -1.2409677548340217e-11 1.3696240783241655e-10
#
SENSITIVITY_A GU.Throughput.1st.fits
#


# 0 order (BEAM B) *******************
BEAMB -85 116
MMAG_EXTRACT_B 30
MMAG_MARK_B 30
# 
# Trace description 
#
DYDX_ORDER_B 0
DYDX_B_0 0.09447092519299671 -1.9602476615012397e-05 5.751452508739299e-05 4.0111749793837556e-10 -4.1768952416968095e-10 -3.6985366991384088e-09
# 
# X and Y Offsets 
#
XOFF_B 0.0
YOFF_B 0.0
# 
# Dispersion solution 
#
DISP_ORDER_B 1
DLDP_B_0 -23000.061614147515 -9.646045283975953e-08 -2.52624295390115e-08 6.333928323209001e-12 -5.6294089195393394e-14 3.3967782507454333e-12
DLDP_B_1 1700.0040048940568 -6.511233807942798e-10 1.033509767020798e-09 -7.893517766318493e-14 -1.9111403091503932e-13 4.654473466209288e-14
#
SENSITIVITY_B GU.Throughput.0st.fits
#


# -1 order (BEAM C) *******************
BEAMC 489 1119
MMAG_EXTRACT_C 30
MMAG_MARK_C 30
# 
# Trace description 
#
DYDX_ORDER_C 1
DYDX_C_0 -1.3745251280636102 -0.0004166620793305387 0.00030781991744431526 3.5625065790500023e-08 -1.015616705292654e-08 -6.809703219697665e-09
DYDX_C_1 0.005259942743965531 2.2710135102429772e-07 -4.832866701253246e-07 -2.9893769156851795e-11 1.539093571694599e-11 2.3835803634126165e-12
# 
# X and Y Offsets 
#
XOFF_C 0.0
YOFF_C 0.0
# 
# Dispersion solution 
#
DISP_ORDER_C 1
DLDP_C_0 19.622863325050346 -0.0013950075981674912 8.290816665678019e-05 1.0549537131994977e-08 -6.884616011479114e-08 3.0531479700463135e-08
DLDP_C_1 4.327391204728536 -2.3365189342607888e-05 -1.2311387785374968e-07 7.0815635974162715e-12 1.1892033820082453e-10 -4.250902118154727e-11
#
SENSITIVITY_C GU.Throughput.-1st.fits
#


# 2 order (BEAM D) *******************
BEAMD -2123 -1030
MMAG_EXTRACT_D 30
MMAG_MARK_D 30
# 
# Trace description 
#
DYDX_ORDER_D 1
DYDX_D_0 -3.7902089554161935 -0.0005938780884699344 0.0004094352465421359 3.528558369624729e-08 8.932646120211718e-09 -4.700678524711011e-09
DYDX_D_1 -0.005334782127730197 -3.4904378900281923e-07 3.8534743707073547e-07 1.4929393517554934e-11 1.3055894861406129e-11 -2.410976643143283e-13
# 
# X and Y Offsets 
#
XOFF_D 0.0
YOFF_D 0.0
# 
# Dispersion solution 
#
DISP_ORDER_D 1
DLDP_D_0 152.5816234883091 0.0024008885842128317 -6.933000876497405e-05 2.14925824016012e-08 -6.282588631094311e-09 5.991593668383653e-09
DLDP_D_1 -2.15372444886087 1.8139334009557958e-05 -2.4736490012643057e-07 -7.297348281711162e-11 -5.403947413929594e-12 1.3050911678919327e-11
#
SENSITIVITY_D GU.Throughput.2st.fits
#


# -2 order (BEAM E) *******************
BEAME 1079 2193
MMAG_EXTRACT_E 30
MMAG_MARK_E 30
# 
# Trace description 
#
DYDX_ORDER_E 1
DYDX_E_0 -5.5518511762976965 -0.0002398046719970402 0.00045376018467024545 1.2301023991565204e-08 7.2902255958184095e-09 -4.102819822051389e-09
DYDX_E_1 0.008329905249142448 -1.8413918879925165e-08 -5.436656649381185e-07 -2.6646238487129985e-12 -3.709762742803416e-14 -7.605203781010001e-13
# 
# X and Y Offsets 
#
XOFF_E 0.0
YOFF_E 0.0
# 
# Dispersion solution 
#
DISP_ORDER_E 1
DLDP_E_0 202.1473421947678 -0.004975140846853297 0.00016529623413855262 1.4193654115130447e-07 -4.0450109813491415e-08 8.055194714392919e-09
DLDP_E_1 2.0094320893783557 -7.885520550681949e-06 2.2521694507180684e-08 -6.827627649913586e-11 2.20617900878291e-11 -6.569233565412959e-12
#
SENSITIVITY_E GU.Throughput.-2st.fits
#

