INSTRUMETN CSSTSLS
GRATING GV
WAVELENGTH 4000 6200

# 1 order (BEAM A) *******************
BEAMA 515 1177
MMAG_EXTRACT_A 30
MMAG_MARK_A 30
# 
# Trace description 
#
DYDX_ORDER_A 1
DYDX_A_0 0.5683725695015461 0.0002625991176114534 7.006920002719378e-05 -5.3902702638251205e-08 2.3077039979626143e-09 2.807448790238151e-09
DYDX_A_1 -0.0016501446166726679 -1.6314360677144785e-07 -2.94319119237387e-07 3.599554493527195e-11 7.82841509626936e-12 -5.649214159083351e-12
# 
# X and Y Offsets 
#
XOFF_A 0.0
YOFF_A 0.0
# 
# Dispersion solution 
#
DISP_ORDER_A 1
DLDP_A_0 185.3546461019509 -0.000553739486924042 5.3805915890072636e-05 -5.202187060025013e-08 3.482547058657821e-08 -1.0439402910731553e-08
DLDP_A_1 5.8616055291181715 7.051466695382349e-05 4.4323240699707057e-07 8.826984506498909e-10 -5.919091363569429e-11 -6.797891528404601e-11
#
SENSITIVITY_A GV.Throughput.1st.fits
#


# 0 order (BEAM B) *******************
BEAMB -121 80
MMAG_EXTRACT_B 30
MMAG_MARK_B 30
# 
# Trace description 
#
DYDX_ORDER_B 0
DYDX_B_0 -0.17005580833505646 0.00026880499899771494 -2.3798013059529185e-05 -5.7760719086943495e-08 1.3962410140423873e-08 -6.620206122560595e-09
# 
# X and Y Offsets 
#
XOFF_B 0.0
YOFF_B 0.0
# 
# Dispersion solution 
#
DISP_ORDER_B 1
DLDP_B_0 -45999.09774226991 3.1551430357268373e-09 -1.3794788501096134e-07 -1.3021428541250755e-11 1.2976075821878232e-11 1.1869104334095063e-11
DLDP_B_1 -2499.9560089978613 4.271751500393071e-09 1.5076141040608468e-09 1.786162770424024e-14 -6.91349749868267e-13 1.0905053935734857e-13
#
SENSITIVITY_B GV.Throughput.0st.fits
#


# -1 order (BEAM C) *******************
BEAMC -1228 -560
MMAG_EXTRACT_C 30
MMAG_MARK_C 30
# 
# Trace description 
#
DYDX_ORDER_C 1
DYDX_C_0 0.8380063286543387 -0.00013037357340758146 0.00010789381564376383 2.3645120613756304e-08 7.177443570652976e-09 8.640448820419748e-11
DYDX_C_1 0.001567576922469369 -2.3588103401500962e-07 2.587174804138313e-07 5.237611819433522e-11 8.373180929474428e-12 2.440567273421944e-12
# 
# X and Y Offsets 
#
XOFF_C 0.0
YOFF_C 0.0
# 
# Dispersion solution 
#
DISP_ORDER_C 1
DLDP_C_0 -10.423786119846984 -0.0003049133625030083 -0.00011551611951060784 1.8026603612641115e-09 8.070120986099926e-08 8.941181458147948e-10
DLDP_C_1 -5.776562119997951 -6.22509833925811e-05 7.038107392176707e-07 -6.686229687154615e-10 9.821045078860279e-11 -7.806501207353245e-11
#
SENSITIVITY_C GV.Throughput.-1st.fits
#


# 2 order (BEAM D) *******************
BEAMD 1163 2343
MMAG_EXTRACT_D 30
MMAG_MARK_D 30
# 
# Trace description 
#
DYDX_ORDER_D 1
DYDX_D_0 2.6368053211882767 -0.0001487917249764907 0.0003703622623883897 1.9895309598789077e-08 1.0363018736288482e-08 3.5771486782834076e-09
DYDX_D_1 -0.0031312799709676325 1.4021043782864998e-07 -5.24123009262728e-07 -2.1478574034804025e-11 -1.7503729317616454e-12 -3.106713845835354e-12
# 
# X and Y Offsets 
#
XOFF_D 0.0
YOFF_D 0.0
# 
# Dispersion solution 
#
DISP_ORDER_D 1
DLDP_D_0 329.969638325766 -0.0035813360418773046 0.00011198209649146591 1.2411351214666428e-07 2.7517115122967678e-08 -2.606855182813427e-08
DLDP_D_1 2.7565938481842873 3.540538606208951e-05 -1.2220260192351779e-08 3.8190045996197453e-10 -2.1688775421445923e-11 -2.713028670441413e-12
#
SENSITIVITY_D GV.Throughput.2st.fits
#


# -2 order (BEAM E) *******************
BEAME -2425 -1222
MMAG_EXTRACT_E 30
MMAG_MARK_E 30
# 
# Trace description 
#
DYDX_ORDER_E 1
DYDX_E_0 2.6839658293922617 0.00010658062025562498 0.000433052445346869 -3.167531570342423e-08 2.9387770353177513e-09 3.446598864135609e-09
DYDX_E_1 0.002792405633393311 4.4178354232831405e-08 4.966573928541092e-07 -6.792686687285308e-12 5.012957092122308e-12 2.5444220795607283e-12
# 
# X and Y Offsets 
#
XOFF_E 0.0
YOFF_E 0.0
# 
# Dispersion solution 
#
DISP_ORDER_E 1
DLDP_E_0 296.38553192902395 0.003649536801750414 -0.0002230800602578193 -4.240677855053282e-08 -6.824297028707603e-08 4.512670565455297e-08
DLDP_E_1 -2.6758122645776115 -2.580594312218447e-05 7.102911247400672e-08 -3.395864979066543e-10 -3.812552080906051e-11 1.3970093723919844e-11
#
SENSITIVITY_E GV.Throughput.-2st.fits
#

