INSTRUMETN CSSTSLS
GRATING GV
WAVELENGTH 4000 6200

# 1 order (BEAM A) *******************
BEAMA -1175 -526
MMAG_EXTRACT_A 30
MMAG_MARK_A 30
# 
# Trace description 
#
DYDX_ORDER_A 1
DYDX_A_0 3.7811150005170577 -0.00040847550433675925 -6.842033189326553e-05 1.8619757183688087e-08 3.603661860143062e-08 -5.031497217905239e-09
DYDX_A_1 0.0030770173661212987 3.073171639924094e-07 8.019425706472434e-08 -2.7948612467844153e-11 3.942101563769058e-11 -5.762799940597462e-12
# 
# X and Y Offsets 
#
XOFF_A 0.0
YOFF_A 0.0
# 
# Dispersion solution 
#
DISP_ORDER_A 1
DLDP_A_0 109.94382316464947 0.007868235003365306 0.0015065145750852876 -3.573800000969111e-07 -1.0268039017966188e-07 -7.850460755067828e-08
DLDP_A_1 -6.453515314520644 6.38479632564617e-05 1.9395165901074966e-06 -4.986927243402116e-10 -6.686889606632738e-11 -8.802275576082156e-11
#
SENSITIVITY_A GV.Throughput.1st.fits
#


# 0 order (BEAM B) *******************
BEAMB -85 116
MMAG_EXTRACT_B 30
MMAG_MARK_B 30
# 
# Trace description 
#
DYDX_ORDER_B 0
DYDX_B_0 0.25307088901743824 -6.180557626035028e-05 -5.6000734874910896e-05 3.008373713368942e-09 9.845541416832335e-09 -2.9588298008839754e-09
# 
# X and Y Offsets 
#
XOFF_B 0.0
YOFF_B 0.0
# 
# Dispersion solution 
#
DISP_ORDER_B 1
DLDP_B_0 -33499.316209131066 -4.889569549249018e-07 -2.3427126649711028e-07 2.7368330807197878e-11 4.086639320943341e-11 -7.796329733737077e-12
DLDP_B_1 2499.95591720129 1.8425804014217365e-08 8.78838126445331e-09 -9.553854423962737e-13 -7.620927232901717e-13 -3.8256078405696343e-13
#
SENSITIVITY_B GV.Throughput.0st.fits
#


# -1 order (BEAM C) *******************
BEAMC 563 1219
MMAG_EXTRACT_C 30
MMAG_MARK_C 30
# 
# Trace description 
#
DYDX_ORDER_C 1
DYDX_C_0 2.6566029956834267 -0.0001993727960878793 0.00013478205560089592 2.20019072660902e-08 5.981916411639331e-10 -3.5067244442090692e-09
DYDX_C_1 -0.004039532789408107 -2.9480238320497073e-07 -4.7165423474910564e-07 1.966361166515235e-11 1.4477239941208906e-11 4.155902211482753e-12
# 
# X and Y Offsets 
#
XOFF_C 0.0
YOFF_C 0.0
# 
# Dispersion solution 
#
DISP_ORDER_C 1
DLDP_C_0 50.87585537884792 -0.004696494578207091 -0.002854838162495125 1.6453343331142687e-07 1.959221511431642e-07 1.4710843212204927e-07
DLDP_C_1 6.20951358391743 -4.939795157297414e-05 2.15357966477588e-06 4.915431616360036e-10 -1.367003148795361e-10 -1.8571831166566514e-10
#
SENSITIVITY_C GV.Throughput.-1st.fits
#


# 2 order (BEAM D) *******************
BEAMD -2331 -1177
MMAG_EXTRACT_D 30
MMAG_MARK_D 30
# 
# Trace description 
#
DYDX_ORDER_D 1
DYDX_D_0 9.833666012049616 -0.0007501727177554814 0.0002824071585063054 5.6696442845204616e-08 6.754806246856363e-09 6.284682472856921e-09
DYDX_D_1 0.007654095586821002 -1.0181243560613835e-08 4.214752513804997e-07 6.842069982346843e-12 3.773424291124153e-12 3.3798332367439925e-12
# 
# X and Y Offsets 
#
XOFF_D 0.0
YOFF_D 0.0
# 
# Dispersion solution 
#
DISP_ORDER_D 1
DLDP_D_0 250.93629876949882 0.007519589214984534 -8.221434996810977e-05 -1.646466862447826e-07 -6.239869545694855e-08 5.310920729533761e-08
DLDP_D_1 -3.068132894666639 3.2728867382170696e-05 2.432956605833829e-07 -2.1317184155101302e-10 -3.349165144997328e-11 3.527019538067007e-11
#
SENSITIVITY_D GV.Throughput.2st.fits
#


# -2 order (BEAM E) *******************
BEAME 1234 2416
MMAG_EXTRACT_E 30
MMAG_MARK_E 30
# 
# Trace description 
#
DYDX_ORDER_E 1
DYDX_E_0 9.093523836228135 -0.0005185506682510039 0.0002952504909479164 3.860624463995974e-08 2.2011624918106087e-08 -9.895919094066933e-10
DYDX_E_1 -0.009402968049811306 9.375727884809776e-08 -5.679892663020363e-07 8.422649782599757e-13 4.763112243232167e-13 -2.1843736228447546e-14
# 
# X and Y Offsets 
#
XOFF_E 0.0
YOFF_E 0.0
# 
# Dispersion solution 
#
DISP_ORDER_E 1
DLDP_E_0 377.8326848831898 -0.008207663120495638 -0.0006468560814309217 1.9910216431424405e-07 7.172846988543768e-09 8.376365525625963e-08
DLDP_E_1 2.832921872688475 -1.7507038752476333e-05 8.357142715248528e-09 8.295967711310619e-11 7.648472344635591e-12 -5.2668572819119176e-11
#
SENSITIVITY_E GV.Throughput.-2st.fits
#

