INSTRUMETN CSSTSLS
GRATING GV
WAVELENGTH 4000 6200

# 1 order (BEAM A) *******************
BEAMA 517 1176
MMAG_EXTRACT_A 30
MMAG_MARK_A 30
# 
# Trace description 
#
DYDX_ORDER_A 1
DYDX_A_0 1.85705160718601 0.0004347912168139198 6.925226274438001e-05 -7.674936379751928e-08 2.5360665429182694e-09 1.8413314253112366e-10
DYDX_A_1 -0.005186075377319128 -2.0657847771207873e-07 -1.6583805404903412e-07 4.736933399389434e-11 4.4009367853254e-13 -1.276594051207188e-11
# 
# X and Y Offsets 
#
XOFF_A 0.0
YOFF_A 0.0
# 
# Dispersion solution 
#
DISP_ORDER_A 1
DLDP_A_0 188.5534512025415 -0.0014704692416785 0.000612543977346908 2.007850076738281e-08 1.4194592531170268e-07 -7.759229950390409e-08
DLDP_A_1 5.8710248614458616 6.45250072921093e-05 -1.4118759316469087e-06 6.713375536655811e-10 -6.079099534683966e-11 8.444062529419406e-11
#
SENSITIVITY_A GV.Throughput.1st.fits
#


# 0 order (BEAM B) *******************
BEAMB -121 80
MMAG_EXTRACT_B 30
MMAG_MARK_B 30
# 
# Trace description 
#
DYDX_ORDER_B 0
DYDX_B_0 -0.6502362516331273 0.0004126085974983609 5.9238627024131765e-05 -6.016741597313287e-08 -4.177679752221959e-09 -9.246098631665313e-09
# 
# X and Y Offsets 
#
XOFF_B 0.0
YOFF_B 0.0
# 
# Dispersion solution 
#
DISP_ORDER_B 1
DLDP_B_0 -45999.098088249615 -2.020488673633556e-08 5.832142102590865e-09 1.4235279966501311e-12 -3.2422967887568725e-12 1.847739810041484e-12
DLDP_B_1 -2499.955996231492 3.442384528615149e-09 -3.91231110580748e-09 -8.790575563626698e-13 -1.8010738964679679e-13 3.4116166635686207e-13
#
SENSITIVITY_B GV.Throughput.0st.fits
#


# -1 order (BEAM C) *******************
BEAMC -1224 -562
MMAG_EXTRACT_C 30
MMAG_MARK_C 30
# 
# Trace description 
#
DYDX_ORDER_C 1
DYDX_C_0 1.9189304176461794 0.00016437384653731756 0.00024569314445877714 -1.7652527457023917e-08 4.451536629733484e-09 -1.8335218213849877e-08
DYDX_C_1 0.00413376823690816 4.0839136662136974e-08 3.196775619036203e-07 2.0152360067018585e-11 5.862317721933411e-12 -9.936558869234138e-12
# 
# X and Y Offsets 
#
XOFF_C 0.0
YOFF_C 0.0
# 
# Dispersion solution 
#
DISP_ORDER_C 1
DLDP_C_0 -20.912091752582096 0.002162826416752504 0.001190637315609984 -5.0299184321331906e-08 -2.0693196648130572e-07 -3.3965803556694406e-08
DLDP_C_1 -5.8016139605407915 -5.3551336327594006e-05 8.780780945442022e-07 -6.738910157909035e-10 -1.3621195992295904e-10 1.3440892596250982e-11
#
SENSITIVITY_C GV.Throughput.-1st.fits
#


# 2 order (BEAM D) *******************
BEAMD 1168 2342
MMAG_EXTRACT_D 30
MMAG_MARK_D 30
# 
# Trace description 
#
DYDX_ORDER_D 1
DYDX_D_0 7.018126896585679 0.00038481511440685416 0.0004494806932104142 -1.0419441298036432e-07 -1.112372803767778e-09 -5.1808602094761285e-09
DYDX_D_1 -0.008748895260191009 -5.149010051129555e-08 -5.215894051986582e-07 3.429591843624662e-11 2.372308962315323e-12 -2.0762709560078255e-12
# 
# X and Y Offsets 
#
XOFF_D 0.0
YOFF_D 0.0
# 
# Dispersion solution 
#
DISP_ORDER_D 1
DLDP_D_0 336.1795490128755 -0.0032204504514754106 -0.00016762481213696898 -2.39993516082151e-08 5.3491579150662175e-08 1.965582670807892e-09
DLDP_D_1 2.757082551671474 3.218019411536404e-05 -2.1509338462577187e-07 3.661907345086239e-10 -1.1815103338002076e-11 9.492832346507823e-13
#
SENSITIVITY_D GV.Throughput.2st.fits
#


# -2 order (BEAM E) *******************
BEAME -2418 -1225
MMAG_EXTRACT_E 30
MMAG_MARK_E 30
# 
# Trace description 
#
DYDX_ORDER_E 1
DYDX_E_0 7.051618660592987 0.00044993052313633966 0.000568576071753248 -9.125941476168853e-08 -1.0197118771739255e-08 -7.464099958501876e-09
DYDX_E_1 0.007720397670144312 2.6306139125746e-07 5.241981337161832e-07 -3.5736030680706604e-11 -2.277967960677812e-12 1.0174827866049466e-12
# 
# X and Y Offsets 
#
XOFF_E 0.0
YOFF_E 0.0
# 
# Dispersion solution 
#
DISP_ORDER_E 1
DLDP_E_0 289.31403069472975 0.004047957141301616 -3.82568269975507e-05 -6.349616609463875e-08 -8.610204084767976e-08 3.4060345226442515e-08
DLDP_E_1 -2.6880282068614423 -2.2955090390629592e-05 5.096417600598682e-09 -2.9431409350852013e-10 -2.022240038595323e-11 2.7697301839390107e-11
#
SENSITIVITY_E GV.Throughput.-2st.fits
#

